|Voltage Detection Switch Low||Voltage Detection Switch High|
|Switch Setting||20V||150V||550V||1000V||Switch Setting||20V||150V||550V||1000V|
|Low 1||–||–||–||–||High 1||–||150mm||450mm||775mm|
|Low 2||0||High 2||150mm||550mm|
|Low 3||20mm||High 3||550mm|
|Low 4||40mm||High 4||600mm|
|Low 5||65mm||High 5||600mm|
|Low 6||100mm||300mm||High 6||200mm||625mm||900mm|
|Low 7||40mm||260mm||High 7||710mm|
|Low 8||410mm||High 8||750mm|
|Low 9||180mm||475mm||High 9||5mm||770mm|
|Low 10||525mm||High 10||10mm||775mm|
|Low 11||525mm||High 11||10mm||825mm|
|Low 12||180mm||575mm||775mm||High 12||10mm||310mm||750mm||1100mm|
SWER and Pole test results refer to: Approach to Pole, with energised reference points.
These calibration figures can vary due to varying weather conditions, but it will be a safe guide on pole approach, that the further away the GLM Mini alerts the higher the leakage that is present.