+61 2 4952 6304
Detects electromagnetic and AC electrical voltage sources from 110 volts to 750,000 volts at a range of 200mm to 300mm.
View Modiewark Contact and Non-Contact Voltage Tester
The new G.L McGavin Digital hand held Proximity Tester FR30 is designed for Electricians, Technicians and the Fire & Rescue departments.
View FR30 Rescue Tester
Non-contact voltage detector for use by persons approaching metal clad meter boxes in residential or commercial situations.
View Mini Meter Reader Voltage Tester
Pocket-sized non-contact power pole tester to aide in the identification of cracked and broken insulators.
View Mini SWER & Pole Voltage Tester
The GLM MIni Meter Reader Features a full function internal self testing unit, low battery indicator detection and unit activation indication, providing a safe and reliable instrument.
The battery compartment is located on the rear of the tester and requires a 9 volt pp3 battery. There is an option of a rechargeable battery for some models.
Ensure the battery is fitted the correct way placing the positive terminal as indicated.
When the unit is activated a GREEN indicator light is illuminated stating the unit is on. This indicates a good battery connection only.
MR-push-light To the bottom right of the GLM MINI Tester, an Orange indicator light will appear when battery levels reach 8.2 volts, this indicates the battery should be replaced or recharged. As the battery output lowers below this point the tester will continually show activation without turning off. (See: battery replacement)
The self test button located on the front face of the GLM MINI is a full function test of all the components of the unit.
This feature should be activated every time the unit is turned on and at regular intervals during the working day. In the event the tester fails please refer to (service and trouble shooting).
This switch controls the sensitivity of the GLM MINI.The high setting will allow detection of high voltages 7,000 volts to ground outdoors 15 metres away and detect 240 volts 1 metre away.The low setting will change the detection distance to 1 metres away at 7,000 volts to ground outdoors and detect 240 volts inside at 10 mm away from an active GPO.